LabPac® Solvent Extract 75i
Pre-mixture 75% IPA, 25% De-Ionized H20 (1 Liter)
Amount used per OmegaMeter 30 - 42 liters (8 - 11 gallons), depending upon test cell size.
The OmegaMeter uses a wash solution of 75% by volume of ACS reagent grade isopropyl alcohol (IPA) and 25% by volume of deionized distilled water.
This "wash" solution is also referred to as the "Test Solvent" or "Extract Solvent" as its purpose is to wash off, or extract contaminates from a circuit board.
You initially fill your ionic test equipment with this Test Solvent.
The solution concentration in the equipment (Alcohol/Water Ratio) may eventually change due to evaporation and the level may drop due to drag on the tested assemblies.
You will use 99% IPA and pure deionized water to adjust the concentration when necessary to bring it back into compliance. The ionic test equipment generally has a menu-prompted procedure to accomplish this.
Once the concentration is brought back to the 75%/25% compliant level, then use this pre-mixed Test Solvent to bring the level of fluid to the full mark.
Material Safety Data Sheet (MSDS) for the Solvent Extract Solution (pdf)
This product was added to our catalog on Monday 22 March, 2021.